|Published (Last):||2 June 2008|
|PDF File Size:||2.75 Mb|
|ePub File Size:||17.62 Mb|
|Price:||Free* [*Free Regsitration Required]|
Check the supply voltage is correct before connecting the instrument. The instrument must not be opened. FT1 in parallel with inductive loads could be useful.
It is provided with four relay outputs to control compressor, defrost - which can be either electrical or hot gas - the evaporator fans and the lights. It is also provided with three NTC probe inputs, one for temperature control, one to control the defrost end temperature of the evaporator and the third, optional, for the display. There are two digital inputs free contact for the door switch and configurable by parameter.
In case of fault in the thermostat probe the start and stop of the compressor are timed through parameters "COn" and "COF". The compressor operates in continuous mode for the time set through the "CCt" parameter. The cycle can be terminated before the end of the set time using the same activation key, 3. At the end of defrost the drip time is controlled through the "Fdt" parameter.
This can be used to make sure circulation of air only if his temperature is lower than set in "FSt". By holding it pressed for 3s when max or min temperature is displayed it will be erased.
Dixell WING XW60L Installing And Operating Instructions