This creates versatility on the bench and may be all the source you ever need. In the high-resolution mode, you get bit resolution with ppm, day accuracy. The low-resolution mode provides bit resolution with ms settling times. In the low-resolution mode, there are seven ranges. In current, there are four ranges of output, from 0. Accurate AC Generate ac voltage outputs, including sine, triangle, and square waves, at frequencies of up to 1MHz.
|Genre:||Health and Food|
|Published (Last):||4 November 2011|
|PDF File Size:||4.92 Mb|
|ePub File Size:||19.40 Mb|
|Price:||Free* [*Free Regsitration Required]|
Fell We want to keep all magnetic and temperature variations away from sensitive LTZ reference. Two opposite DIN -type connectors are used on each card, forming some kind of interlocking connection. I have idea to make low-gain divider opamp board, so Maanual could generate lower voltages more accurately, as high-resolution ranges are only 1V and 10V.
Learn more — opens in new window or tab. There is free space in unit and few extra chassis mounting studs, which are just asking for extra stuff to be mounted. They have hundreds of greatly remastered manuals, so be sure to check their repository for other goodies. For ,anual information, see the Global Shipping Program terms and conditions — opens in a new window or tab.
Front panel board is exactly the same as A7 from A. Learn More — opens in a new window or tab Any international shipping is paid in part to Pitney Bowes Inc. Piece of 4mm rubber pad manua holding everything in place securely, screwed down to existing two metal stubs on chassis. HP A Function Generator. In many cases an old unit might have been retrofitted with newer firmware, so later firmware revision cannot be used as proof of the date of manufacture.
Sell now — Have one to sell? Contents Intro Disclaimer Initial inspection and basic disassembly Manuals references Repair workflow Boards condition check Diagnostics Firmware Tweaks Main reference upgrade Additional output amplifier for 0.
Back to home page. Interest will be charged to your account from the purchase date if the balance is not paid in full within 6 months. Please enter 5 or 9 numbers for the ZIP Code. HP A Signal Generator. This seller is currently away until Jan 02, All output and interboard connections are done with Teflon insulated coax wire, which is required by ability of sourcing high voltages with frequencies up to 1MHz.
People who viewed this item also viewed. Overall internal condition I would mark as Aeverything very clean, metal chassis is shiny and one could easily say this unit is just got out from factory. If you would like to support us by making a donation please click the Paypal button: Also spotted unsoldered cap, likely slipped thru QC: Select a valid country. As usual, all photos are clickable for mannual version.
This hybrid utilizes a beryllia BeO substrate, thick film resistors, ceramic capacitors and semiconductor chips to maximize reliability, minimize size and give top performance.
All information posted here is hosted just for education purposes manuaal provided AS IS. To generate single binary I used simple python snipped like this. Refer to eBay Return policy for more details. Learn More — opens in a new window or tab International shipping and import charges paid to Pitney Bowes Inc.
Read more about the condition. Differences from A featured board:. Pleasant looks and feel during operation guaranteed. Most Related.
Providing the finest test equipment solutions since 1992
In no event shall the author, xDevs. HP A Universal source repair project Well, will see how serious the problem is. It is backlit with red LED so you can see it even without x removal. Mouse over to Zoom — Click to enlarge. Copper foil soldered to ground on power coax wire is located on top. Now, find a best spot for module location.
HP (Hewlett-Packard) HP 3245A User Manual